If the Planmeca device works in PMSample but gives an error in our software stating that the device hasn't been calibrated, go into the SED/IXDLL options and change the resolution from High to Normal.
- To access the SED options in XrayVision 4, click the Tools menu at the top, selecting Hardware Options
Double click on Planmeca
- To access the IXDLL options in DCV or XVCapture, select Advanced User Tools from the main menu, click Preferences and type in the password "AKRON" without quotes, then click OK
Select Imaging Extensions from the left side towards the bottom, click the red carrot (>) next to Digital X-ray Image Capture and select Planmeca
Article Version 1.0 2019.11.1